Skip to main content
Get your brand new Wikispaces Classroom now
and do "back to school" in style.
Pages and Files
About this Wiki
HIPE video tutorials
HIPE on Twitter
Contributing to HIPE
Back to HSC homepage
SPIRE mapmaking and photometry with HIPE 11
8 Nov. 2013, 17:30
Chair: Luca Conversi
In the call: Luca Conversi, Ivan Valtchanov, Ram Kesh, Silentrash, vero, Javier Diaz, Alvar García, Bruno Merín
Agenda for the webinar:
Effect of the updated Neptune model used in HCSS 11 onward: from ESA2 model to ESA4
SPIRE map-making workshop report summary
Description of new level 2.5 and 3
How to reprocess the cross-calibration with Planck for parallel mode observations (so-called new level 2.5)
For the second to last point, we plan to run a demo script (attached below). Needed setup:
Planck HFI-545 and HFi-857 maps available here
on the HSC FTP area
The attached demo script:
And here is the PDF of the presentation:
HIPE 11 SPIRE Photometer Updates.pdf
Video-recording of the webinar:
Contents of the webinar:
Minute 0:00 : Effect of the updated Neptune model used in HCSS 11 onward: from ESA2 model to ESA4
Minute 6:50 : Description of new level 2.5 and 3
Minute 10:25 : SPIRE map-making workshop report summary
Minute 19:07 : Demo script introduction
Minute 23:05 : How to reprocess the cross-calibration with Planck for parallel mode observations (so-called new level 2.5)
Minute 38:20 : Q&A
Questions and answers:
Q - Silentrash: which map maker should we use?
A - Luca Conversi: there is no such a thing as a mapper that works for every observations, but slide 6 of the presentation gives you the advantages of each of them and a link to the full report discussing the test comparison.
Q - Silentrash: can both all map makers used for point sources and for extended sources.
A - Luca Conversi: Yes, although for doing photometry the user might need to make a few further post-processing to the images before he/she can do e.g. aperture photometry. A whole description of how to do photometry in SPIRE scan maps was presented in the previous webinar of this series (
Q - Bruno Merín: can you use Planck data directly from the Planck Science Archive to run this demo?
A - Luca Conversi: Not directly from the archive. The simplest reason is that the full resolution Planck FITS file from the archive have different structure from those in the archive (HDU_x's have different layout). There is also another scientific reason: the 500 and 350 micron have non-gaussian beams of 4.5-4.7 arcmin, so we needed to convolve them to a 8 arcmin gaussian beams. Tests with the full-res Planck data are not yet being successful. The demo script just calculates an off-set constants for each band.
help on how to format text
Turn off "Getting Started"